Customer Senvice Hotline
Mechanical scale: use and description of mechanical scale
Use and description of mechanical scale
1、 How to use the mechanical scale:
When using the mechanical scale, the object should be placed on the load as lightly as possible, and should not be placed with force. And after each use, the parts are removed to avoid blade damage.
If the mechanical platform scale is used continuously, whether the empty scale is balanced after a certain number of times to ensure the accuracy of measurement.
In the process of using the mechanical help scale, except for calibrating the empty scale, nothing can be hung on the frame, otherwise the measurement accuracy of the scale will be affected.
In order to avoid damaging the mechanical scale in use, it is necessary to pay attention to the large weight of the measured object, which shall not exceed the weighing indication of the mechanical scale.
2、 Precautions before using the mechanical scale:
When using, first pack the inside and outside of the mechanical scale, and place it on a flat and hard ground, with four wheels touching the ground at the same time to prevent it.
Measurement of the influence of adult orientation.
When using the scale, shake the bearing plate to ensure that each blade of the scale is well connected with the knife. At the same time, check whether the measuring rod is deformed. Then, move the weight point ring and measuring rod of the support (it will move to the "D" position). If it is not balanced, turn the adjusting screw to balance the empty scale.
3、 Maintenance of mechanical scale:
When the platform scale is used for a long time, each blade and tool carrier ring will be subject to certain wear and fatigue, and the accuracy and sensitivity will reduce each other. Therefore, the mechanical scale must be checked and calibrated regularly. Generally, it should be overhauled in the second half of the year or every year, and the increase should be verified at least once a year.
The collimator shall be kept dry and clean at all times to prevent estimated liquid substances and dirt from remaining in the collimating frame and tool holder. The lifting ring of the scale should always be kept clean so as not to affect the sensitivity (it should also be added carefully to avoid losing the original accuracy).
It shall be kept intact and shall not be unloaded at will.
During transportation, avoid collision. Never let the lever lift the measuring lever, otherwise the scale body and parts will be damaged, the measuring lever will be deformed, and the accuracy of the mechanical scale will be affected.
Do not put the mechanical scale in the open air, and do not put scale and corrosive articles together, so as to avoid component corrosion affecting the service life.
Nanning Weighing Instrument Factory was founded in 1956. Its main products include weights, scales, weights, mechanical scales, mechanical scales, standard weights, old scales, scale accessories, mechanical scale accessories, etc. It has a long history of more than 50 years of production and operation of weighing instruments. The registered capital is 2.82 million yuan, the total assets are 13 million yuan, the floor area is 7280 m2, and the building area is 11358 m2. Annual production capacity: 500 electronic truck scales, 50000 desk scales and 50000 electronic platform scales. It is a professional manufacturer engaged in the R&D, design, production, sales and service of industrial weighing instruments, household weighing instruments and other products, which was established earlier and has a relatively large scale in Guangxi weighing instrument industry.