Customer Senvice Hotline
What are the reasons for electronic weighing data jumping?
Electronic weighbridge is an important weighing equipment for weighing bulk cargo, which is mainly composed of weighing body, weighing sensor and display instrument. As we all know, the electronic weighbridge will fail for various reasons. Now, our weighbridge manufacturer will analyze a common failure - weighing data jumping. Why do users encounter this situation? This article will summarize the reasons for this.
The main reasons are:
1. The main line connecting the display to the electronic weighbridge is broken.
2. Unexpected water ingress or water vapor into the junction box.
3. The compression capacitor in the display is aging and damaged.
4. The weighing instrument and the main line plug or socket are oxidized during long-term use.
5. The feet of the electronic platform scale are loose or the four feet are unbalanced.
Huizhou Weighbridge Weighing Equipment Co., Ltd., through scientific management, strictly implements the requirements of the quality management system, constantly absorbs the opinions of customers, learns the technical experience of international advanced products, continuously improves the structure and functions of products, and prevents the defects and deficiencies of products, striving to be a good brand in the weighing instrument industry.
We are a professional enterprise integrating research and development, production, manufacturing, sales and service, and providing accessories for all kinds of industrial electronic scales and automatic weighing systems. "Let every customer be satisfied" is our commitment to customers.