Chinese | English
Nanning Weighing Machine Factory
Customer Senvice Hotline
Replace the electronic truck scale instrument, replace the old instrument, and connect the new instrument. Here, be careful not to be busy entering the sensor address and relevant parameters. The correct steps are:
1. Modify the index value and maximum weighing of the instrument;
2. Input sensor address and relevant parameters and ensure accuracy;
3. Adjust the serial port data format and baud rate of the instrument.
4. Restart the instrument, observe the zero point of the instrument, test the eccentric load and weighing of the scale body, and resume use after it is normal.
The above steps have a certain logical relationship and cannot be reversed. According to experience, if the input sensor address and related parameters are modified first, and then the index value and maximum weighing are modified, the instrument display will seriously deviate from the zero point. If the input of sensor parameters is incorrect, the unbalanced load of the scale will be very abnormal.
> Site Map
About us | Honor | News Center | Technical service | Feedback | Contact us
Scale, platform scale business hotline: 0771-3159933
Platform scale, electronic scale business hotline: 0771-3153889After service
Complaint telephone: 0771-3154958
Company address: No. 10, North inter South Road, Nanning.
Q Q： 1363900082
Bus Line: Chaoyang Square can take No. 18; Langdong Station, Xixiangtang Station can take No. 76; Jinqiao Station can take No. 81; Jiangnan Station can take No. 207 and No. 81 to get off at Beijida intersection.
Copyright © Nanning Weighing Machine Factory